- Review Article
- Open Access
Measurement of Physical Activity and Energy Expenditure in Wheelchair Users: Methods, Considerations and Future Directions
Sports Medicine - Open volume 3, Article number: 10 (2017)
Accurately measuring physical activity and energy expenditure in persons with chronic physical disabilities who use wheelchairs is a considerable and ongoing challenge. Quantifying various free-living lifestyle behaviours in this group is at present restricted by our understanding of appropriate measurement tools and analytical techniques. This review provides a detailed evaluation of the currently available measurement tools used to predict physical activity and energy expenditure in persons who use wheelchairs. It also outlines numerous considerations specific to this population and suggests suitable future directions for the field. Of the existing three self-report methods utilised in this population, the 3-day Physical Activity Recall Assessment for People with Spinal Cord Injury (PARA-SCI) telephone interview demonstrates the best reliability and validity. However, the complexity of interview administration and potential for recall bias are notable limitations. Objective measurement tools, which overcome such considerations, have been validated using controlled laboratory protocols. These have consistently demonstrated the arm or wrist as the most suitable anatomical location to wear accelerometers. Yet, more complex data analysis methodologies may be necessary to further improve energy expenditure prediction for more intricate movements or behaviours. Multi-sensor devices that incorporate physiological signals and acceleration have recently been adapted for persons who use wheelchairs. Population specific algorithms offer considerable improvements in energy expenditure prediction accuracy. This review highlights the progress in the field and aims to encourage the wider scientific community to develop innovative solutions to accurately quantify physical activity in this population.
Predicting energy expenditure from physical activity is inherently more challenging in persons with chronic physical disabilities who use wheelchairs due to altered movement patterns and variations in metabolically active muscle mass.
Recent studies have successfully attempted to utilise technological advancements (i.e. multi-sensor devices) to measure physical activity and predict energy expenditure (using population or activity-specific algorithms) in persons who use wheelchairs.
Combining measurement methods (both self-report and objective) might provide greater contextual information about the types and purpose of activities being performed by persons who use wheelchairs. This has implications to inform the wider public health agenda by promoting physical activity and reducing non-communicable diseases in persons with chronic physical disabilities.
Considerable evidence now exists to support the beneficial effects of physical activity (PA) for human health and wellbeing [1–3]. However, the majority of this evidence is from research in adults without disabilities. Our understanding of the impact and importance of PA for populations with chronic physical disabilities, particularly those who use wheelchairs, is therefore lacking. One of the major barriers to the acquisition and analysis of PA data in populations who use wheelchairs is the uncertainty surrounding the validity and reliability of the existing PA measurement tools. Improved assessment of habitual PA would permit; appropriate cross-sectional comparisons to biomarkers of metabolic health allow researchers to comment on the efficacy of behaviour change interventions and potentially inform PA guidelines . This review provides a detailed evaluation of the available tools within the context of their potential application in persons who use wheelchairs. Firstly, three of the most frequently utilised self-reported measures will be described and evaluated. Then our attention will turn to the increasingly employed objective methods of measuring PA.
The PA behaviour of persons who use wheelchairs is inherently difficult to measure due to the heterogeneous nature of the population, whereby different disability aetiologies responsible for the use of a wheelchair result in highly variable movement patterns. Common physical disabilities that require prolonged use of a wheelchair include spinal cord injury (SCI), amputation, multiple sclerosis, cerebral palsy and cerebrovascular disease. Currently, it is problematic to accurately equate PA into units of energy expenditure (EE), as EE varies significantly from person to person depending on body mass, type of physical disability and efficiency of movement. Due to movement being primarily restricted to the upper body, the energy cost of most exercise and activities of daily living performed by persons who use wheelchairs result in a considerably lower energy cost (−27%) than those reported in the general population [5, 6]. The smaller skeletal muscle mass activated to perform certain activities does not achieve the same whole-body metabolic rate. Metabolic equivalents (METS) are often used to express the energy costs of PA as multiples of resting metabolic rate (RMR) . However, the conventional MET value (oxygen uptake of 3.5 mL kg−1 min−1) is not applicable for persons with a disability, as disuse/paralysis results in atrophy of leg fat-free mass (FFM) [8, 9]. RMR is influenced by FFM , which explains why RMR is reduced in persons with disabilities who use wheelchairs compared to adults without disabilities . For example, commonly used equations to predict RMR in persons with spinal cord injury (SCI) overestimate measured requirements by 5–32% . Considering RMR is the largest component of total daily energy expenditure (TDEE) (up to 80% for sedentary individuals ), error in the prediction of this component using existing algorithms for persons without disabilities can have profound implications for accurately predicting TDEE. Consequently, approaches that solely measure physical activity energy expenditure (PAEE) might have greater utility, particularly as this is the most malleable component of TDEE.
The PA monitoring field is evolving at a rapid pace. However, the development of validated self-report and objective tools to quantify PA/EE in persons who use wheelchairs remains relatively under researched. It is not always feasible to use criterion methods (i.e. indirect calorimetry, observation, doubly labelled water) to measure free-living PA/EE, as these techniques require expensive/sophisticated equipment or are impractical for use outside of the laboratory. Therefore, this review provides an overview of the predominant methods of measuring PA/EE in persons who use wheelchairs. Specifically, we describe and review the different self-report and objective tools currently available whilst also considering their potential strengths and limitations.
Until recently, the quantification of free-living PA in persons who use wheelchairs had been reliant on outputs from self-report measures [14, 15]. Self-report questionnaires offer researchers an inexpensive and easy-to-administer method of measuring PA. However, these methods are reliant upon the accuracy of the participants’ memory and recall. Furthermore, it has been suggested that self-report measures fail to adequately quantify the lower end of the PA continuum [16, 17], suffer from floor effects (lowest score is too high for inactive respondents) and participant over-reporting . Besides these general limitations, specific issues pertaining to the administration of the three predominant questionnaires (Table 1) used to predict components of PA in this population are discussed below. It is noteworthy that not all were developed and/or validated for persons who use wheelchairs.
The Physical Activity and Disability Survey (PADS)  was one of the first questionnaires developed but was validated for participants with a wide range of disabilities ranging from stroke to type-2 diabetes, and subsequently, a revised version (PADS-R) in persons with neurological conditions [20, 21]. Therefore, it could be argued that the content of the PADS fails to capture activities specific to the lifestyle of persons that use wheelchairs. The Physical Activity Scale for Individuals with Physical Disabilities (PASIPD)  was adapted from the Physical Activity Scale for the Elderly (PASE) and follows a similar format to that of the International Physical Activity Questionnaire (IPAQ) . Despite being developed in people with both visual and auditory disabilities, its implementation in people with locomotor impairment and SCI means it could be considered sensitive to persons who use wheelchairs. However, only the Physical Activity Recall Assessment for People with Spinal Cord Injury (PARA-SCI) was specifically developed and evaluated for people with SCI.
A distinguishing feature between the three questionnaires is the resource demand to complete each tool (Table 1). The PARA-SCI was designed as an interview-based questionnaire that collects rich behavioural data. Thus, the PARA-SCI is resource intensive because it was developed, as a research tool, to be used in epidemiological studies. For example, it can take between 20–45 min to complete, the cost of the interviewer needs to be considered and there is considerable participant demand. Ullrich et al.  also suggested that the use of the PARA-SCI might have limited application for other investigators, besides the developers, due to the exclusion of subjective appraisals and the technical complexity of interview administration. These limitations were acknowledged by the authors who subsequently developed a new questionnaire to address these limitations. The Leisure Time Physical Activity Questionnaire for People with Spinal Cord Injury (LTPAQ-SCI)  is a brief (5 min) self-report questionnaire specifically designed for persons with SCI that measures minutes of mild, moderate and heavy-intensity leisure time physical activity (LTPA) performed over the previous 7 days but is not capable of measuring other activities of daily living.
Reliability and Validity
The test-retest reliability of the three questionnaires has been examined; however, the PADS has had no reliability studies conducted in persons who use wheelchairs. Therefore, it remains unclear whether the PADS can be reliably used as a measure of physical activity behaviour in this population. A test-retest reliability correlation of .77 was established for the PASIPD in a study of 45 adult patients with a range of disabilities, but these patients did not use wheelchairs . The PARA-SCI is the only instrument tested for reliability in a sample solely consisting of persons who use wheelchairs. To establish the test-retest reliability of the PARA-SCI, 102 people with SCI completed the instrument on two separate occasions a week apart . Intra-class correlations revealed good test-retest reliability for total cumulative activity (.79). However, moderate-intensity (LTPA) and heavy-intensity (lifestyle activity) demonstrated poor levels of reliability (ICC = .45 and .56, respectively).
Establishing the validity of questionnaires is important to ensure that the tool effectively measures what it intends to (i.e. the activity of persons that use a wheelchair). Manns and colleagues  revealed a significant moderate relationship between scores on the PADS and V̇O2 max (r = 0.45). Likewise, comparison of scores from the PASIPD with indicators of physical capacity revealed weak to moderate relationships (V̇O2 max; r = 0.25, manual muscle test; r = 0.35) . However, we contend that equating self-reported PA to physical capacity, rather than a criterion measure of PA, may not be the most appropriate way to ascertain concurrent validity. Measures of physical capacity can be related to numerous variables beyond the users’ PA level.
Results from validity studies indicate that of the three questionnaires, the PARA-SCI has the strongest relationships with criterion measures. During the development and evaluation of the PARA-SCI , criterion (V̇O2 reserve) values displayed a very large correlation with cumulative (LTPA plus lifestyle) activity data (r = 0.79). When data was coded for intensity of activity, large to very large positive correlations were seen for moderate-intensity (r = 0.63) and heavy-intensity (r = 0.88) activity. However, this relationship was weak and non-significant for low-intensity activities (r = 0.27) and consequently, the PARA-SCI under-reported time spent doing activities of low intensity by 10%. Therefore, although these findings indicate some evidence of convergent validity, the results also highlight limitations of self-report measures.
A distinguishing feature between the three disability questionnaires is how they gather information pertaining to the intensity of activity conducted. Evidence from adults without disabilities would suggest superior reductions in mortality risk with vigorous-intensity PA in comparison to light-to-moderate intensity PA [30–32]. Therefore, failure to consider individual differences in PA intensity makes it difficult to detect relationships between lifestyle activities and health outcomes . The PADS employs a single item to examine the overall intensity of structured activity but does not assess the intensity of leisure time activities. A fundamental limitation of the PASIPD is the use of standard MET values as a measure of activity intensity regardless of the participant’s type of disability. If MET values are to be used, it will be necessary to develop a new empirically based supplement to the compendium of physical activity appropriate for persons that use wheelchairs . The inability of the PASIPD and PADS to effectively measure activity intensity prompted the development of the PARA-SCI. Subsequently, the authors of the PARA-SCI conducted a systematic process to develop definitions of three different exercise intensities (i.e. mild, moderate and heavy) specifically for people with SCI . The empirical development of intensity-based definitions suggests the PARA-SCI may be the most effective self-report questionnaire at measuring the intensity of PA in persons with SCI. However, it should be noted that even with such a rigorous development of intensity definitions, the PARA-SCI is still dependent upon the accurate recall of behaviour. Research has also challenged the use of psychophysiological indexes as a measure of perceived exertion in persons with SCI . This could have implications for the prediction of activity intensity using self-report measures in persons with disabilities, which could be influenced by secondary conditions such as chronic pain and discomfort, coupled with the inability to engage large muscle groups in constant rhythmic activities.
Objective sensors overcome many of the shortcomings of self-report methods, predominantly by removing the subjective recall element. The next section will discuss the use of these objective sensors in wheelchair users.
Accelerometers or movement sensors report their outcomes in ‘activity counts’ per unit time or epoch, which are the product of the frequency and intensity of movement. Accelerometers are therefore capable of providing temporal information about specific variables such as the total amount, frequency and duration of PA . They can also monitor the accumulation of moderate-to-vigorous intensity PA (MVPA) and/or sedentary behaviour thanks to the development of population-specific cut-points for activity counts per minute. Despite enormous differences in signal processing and internal components, all accelerometers have similar fundamental properties defined by accuracy, precision, range and sensitivity and should be compared against criterion measurements to demonstrate validity . Monitors have been compared to a selection of criterion laboratory measurements in persons that use wheelchairs: oxygen uptake (V̇O2) [37–39], EE [40, 41] and PAEE [42, 43] measured by indirect calorimetry (Table 2). Studies have utilised different commercial monitors, worn at various locations, validated using diverse activity protocols including propulsion on a wheelchair-adapted treadmill, wheelchair ergometer or over ground, arm-crank ergometry (ACE) and various activities of daily living. Two fundamentally different varieties of accelerometers are widely used in PA research, uniaxial and, increasingly, tri-axial. Uniaxial accelerometers register movement in the vertical axis only, whereas tri-axial accelerometers register movement in the anteroposterior (X), mediolateral (Y) and vertical (Z) axes. In keeping with pooled data from a systematic review of laboratory and free-living validation studies in adults without disabilities , it appears that the greater sensitivity of the tri-axial accelerometer leads to a better prediction of EE than uniaxial accelerometers in persons who use wheelchairs (Table 2).
Monitors Attached to the Wheelchair
Researchers have explored attaching a custom data logger  or biaxial  and tri-axial  accelerometers onto the wheels of a wheelchair. Other preliminary research has simply attached a smartphone (containing a gyroscope and accelerometer) onto the armrest of a wheelchair [48, 49]. Considering the exponential growth of smartphone ownership , this later approach in particular can widely be used to capture certain mobility characteristics such as average speed and distance travelled, functioning in a similar manner to pedometers in persons who do not use wheelchairs. However, despite these approaches being relatively unobtrusive, they are unable to quantify the intensity of activities being performed and are limited in deriving accurate EE estimates. Conger et al.  tried to address this limitation by using a PowerTap Hub attached to the wheel of a wheelchair. The measured hand rim propulsion power explained 48% of the variance in predicting criterion EE. The authors revealed three significant prediction models from this laboratory protocol, with model 3 (incorporating power, speed and heart rate) explaining the greatest variance (87%). Seemingly, the incorporation of a physiological signal significantly improved the prediction of EE.
However, we propose that a device attached to the wheelchair cannot distinguish between self or assisted propulsion, certainly not without complex analyses of the raw acceleration outputs , and are unable to quantify activity out of the wheelchair. Further, it is common for persons who use wheelchairs to have different chairs to participate in various sports or undertake ACE as a mode of exercise. Therefore, a single device attached to a wheelchair will fail to capture moderate-to-vigorous-intensity activity in structured exercise, likely to contribute a large proportion towards TDEE. Moreover, if a person uses a power-assisted wheelchair, signals from devices attached to the chair will provide erroneous measurements regarding upper body PAEE. These limitations need to be considered when using this approach to predict free-living PA/EE in persons that use wheelchairs.
Waist-mounted single-sensor devices, positioned within close proximity to an individual’s centre of mass, have been the mainstay of activity monitoring in cohorts without physical disabilities. Single units worn on the waist can be limited for certain types of upright behaviours that have a low ambulatory component and may involve upper-body work . The measurement error of waist-mounted devices is generally related to the inability to detect arm movements as well as static work (e.g. lifting, pushing, carrying loads). With movement of persons that use wheelchairs predominantly restricted to the upper-body, it is unsurprising that stronger correlations between accelerometer outputs and criterion measurements were reported for devices worn on the upper arm and wrist, r = 0.83–0.93 and r = 0.52–0.93, respectively (Table 2). While two studies [37, 41] have found differences in the strength of correlations between the left and right wrist, these discrepancies could be due to hand dominance or the specific asymmetry of the activities performed in these studies. The predominance of research, however, suggests little to no difference between dominant and non-dominant wrists [38, 39], suggesting freedom/flexibility in selecting either wrist to predict PA/EE in this population.
Combining data from two anatomical locations seemingly does not yield substantial improvements in the strength of correlations or EE estimation error [39, 54]. In some research and development laboratories, accelerometers have been arranged in parallel arrays and positioned at various anatomical locations to monitor the types of activity being performed by postural identification. Such prototype PA monitors were developed to primarily target specific population groups during rehabilitation, including amputees  or inpatients with SCI . Devices with multiple arrays have shown good specificity (92%), agreement (92%) and sensitivity (87%) for the detection of wheelchair propulsion in observational studies . Yet, even when worn for a relatively short period of time, participants self-reported moderate burden . These monitors are relatively obtrusive and, due to reduced memory capacity and battery life, are restricted to short monitoring durations (<48 h). This is not in keeping with current end user requirements of PA monitors. Multi-site prototype arrays are also not typically available outside of the developing laboratory, making validation by other researchers challenging.
A simpler set-up, the Physical Activity Monitoring System (PAMS) , which incorporates a gyroscope-based wheel rotation monitor (G-WRM) and one tri-axial accelerometer attached to the arm or wrist, overcomes the shortcomings of accelerometers attached to the wheelchair alone. When this approach was recently evaluated using a robust laboratory protocol and home-based follow-up session, both the PAMS-arm and PAMS-wrist estimated EE with small biases (M ± E < 10%) . Yet, MAE for predicting EE in persons that use a wheelchair remained elevated (>25%). Kooijmans et al.  also assessed the utility of a tri-axial accelerometer (GT3X+) attached to the wrist and spokes of a wheelchair. However, rater observations reported less agreement (85%) and specificity (83%) for wheelchair propulsion than using multiple-arrays . Whilst less burdensome, disagreement between GT3X+ (Actigraph, Pensacola, FL) outputs and observers was greatest for propulsion on a slope and being pushed whilst making excessive arm movements. Therefore, it is likely that physiological signals, such as heart rate, should be incorporated into the prediction of EE to improve accuracy.
Heart rate (HR) is useful as a physiological variable as it increases linearly and proportionately with exercise intensity and thus oxygen uptake , at least in individuals without disabilities. Keytel et al.  concluded that EE can be accurately predicted from HR after adjusting for age, sex, body mass and fitness. However, during lower intensity PA, there is a weak relationship between HR and EE . This is most likely due to small postural changes causing alterations in stroke volume, or that HR during low intensity PA is affected by external factors such as psychological stress, stimulants, ambient temperature, dehydration and illness . There are a number of ways to use HR data to predict EE, one of the most promising being the FLEX-HR method , which has previously been used in persons with SCI [66, 67]. Despite recent research into the use of various HR indices  and artificial neural networks  in the prediction of V̇O2 in individuals with SCI, it is clear that the accurate prediction of EE using HR is heavily reliant on individual calibration. Hayes et al.  found that the variance in measured EE was considerably improved using an individual calibration (55%) compared to HR alone (8.5%) during five activities of daily living in thirteen individuals with SCI. Considering the type of activities performed, the large variations in cardiovascular fitness and cardiovascular responses to exercise stress persons who use wheelchairs, individual specific HR-EE relationships are necessary for the accurate prediction of EE using HR. This consideration is perhaps even more important for persons with considerable functional impairment or various disability aetiologies that may disrupt the autonomic nervous system, such as high-level SCI (>T6).
New multi-sensor technologies, which include the combination of physiological parameters and accelerometry, have great potential for increased accuracy in assessing EE as they incorporate and minimise the strengths and weaknesses of physiological signals and accelerometry alone. The use of multi-sensor devices has mostly been limited to laboratory based validation of the SenseWear® Armband (SWA) (BodyMedia Inc., Pittsburgh, PA), which is worn on the upper arm, a preferential anatomical location for the prediction of EE in persons that use wheelchairs (Table 2). More detailed components and specifications of this activity monitor have been described elsewhere . It is clear that the proprietary manufacturer’s algorithms intrinsic to the SWA device are not appropriate to predict EE in persons that use wheelchairs, with ICCs < 0.64 [40, 70, 71]. The overestimation of EE by the SWA manufacturer’s model is likely due to the movements typically performed by persons that use wheelchairs (e.g. wheelchair propulsion and ACE) not being included in predefined activity categories. Hence, such activities are misclassified into more strenuous types of PA.
Researchers have developed new EE prediction models (SCI general and activity specific) for the SWA device that have been cross-validated [40, 71]. Where MAE statistics are available [40, 70, 71] weighted means were calculated, with the SCI general (22.7%) and activity specific (18.2%) models performing significantly better than the manufacturer’s model (54.4%). Whilst these findings provide encouragement for the use of the SWA in persons that use wheelchairs with new prediction models, Conger et al.  noticed that even when using the SCI general model, the SWA tended to overestimate EE (27 to 43%), whereas a wrist-mounted accelerometer more accurately predicted EE (9 to 25%) during wheelchair prolusion (Table 3). It is noteworthy that the SWA utilizes upwards of twenty possible output parameters, including heat flux, galvanic skin response and temperature to predict EE. Individuals with high level SCI (>T6) experience impaired thermoregulatory function (reduced sweating response and inability to dilate superficial vasculature ), which might intrinsically effect the error when using SWA in this population. Unfortunately, the acquisition of the company BodyMedia by Jawbone in 2013 resulted in the device being taken off the market and cessation of all BodyMedia web applications. Despite considerable improvements in EE prediction error it seems the future use of this technology is limited.
The Actiheart (Cambridge Neurotechnology Ltd, Papworth, UK) integrates an accelerometer and HR monitor into a single-piece movement monitor. The Actiheart (AHR) unit has been described in detail previously , along with the detailed branched modelling technique it utilises to estimate PAEE through the combination of HR and accelerometer counts . Previous work from our research group  has assessed the performance of this device in a controlled-laboratory environment with a heterogeneous sample of persons who use wheelchairs. Across all activities considerable mean absolute error (MAE) was reported (51.4%) using the manufacturer’s proprietary algorithms to predict PAEE. By using an incremental ACE test, which permitted an individual HR-EE relationship similar to that performed by Hayes et al , individual calibration was incorporated and MAE was considerably reduced to 16.8% across all activities. Individual calibration has also been shown to improve the prediction of EE estimations using this device in free-living [76, 77] and laboratory settings  in adults without disabilities during walking and running. The sizeable improvement in EE prediction error in persons who use wheelchairs with individual calibration may be due to a larger degree of individual variance in cardiovascular function and responses to exercise in this population. Consequently, individual calibration of this monitor is of upmost importance for the accurate prediction of PAEE in persons that use wheelchairs. Furthermore, incorporating individually calibrated HR and acceleration data better captures the differing energy costs of bespoke activities, despite similar acceleration profiles, such as wheelchair propulsion up a gradient or with additional load (e.g. shopping) .
Prediction Accuracy of Methodologies in Free Living Environments
The majority of PA/EE validation research in this population has been performed in a controlled-laboratory environment but there is a paucity of free-living studies (Table 4) primarily due to the practical difficulties or expense associated with ‘gold standard’ EE measurement (DLW). This method is not without limitations; for example minimal information regarding frequency, duration or intensity of activity can be obtained . Furthermore, the estimation of EE is based on the assumption of a mean respiratory exchange ratio (RER) of 0.85, indicative of a standard western diet . Yet, carbohydrate and fat oxidation has been shown to be altered with arm compared to leg exercise  and in paraplegics compared to non-disabled controls . These factors may lead to an increased RER in persons that use wheelchairs, which could violate the assumptions used in the prediction of EE via the DLW technique.
Irrespective of this, Tanhoffer et al.  compared four aforementioned prediction methods (SWA, FLEX-HR, PARA-SCI, PASIPD) to DLW during habitual routines over an extended 14-day period. The authors demonstrated that the two best prediction methods were PARA-SCI and FLEX-HR for both TDEE and PAEE (Table 4). The SWA and PASIPD both performed particularly poorly in the prediction of PAEE, displaying considerable random error as demonstrated by the large 95% limits of agreement. It is noteworthy that the SWA used the aforementioned error-prone manufacturer’s model [40, 70, 71] but could be improved with the SCI general EE prediction model developed by Hiremath et al. . One limitation of the Tanhoffer et al  study is that the length of PA monitoring period for each prediction method varied compared to the criterion method. Total EE collected over a two-week period for the criterion DLW technique was divided by 14 to estimate mean TDEE. However, the objective measures (SWA and FLEX-HR) were only worn ≥12 h on two separate days and subjective measures, the PARA-SCI and PASIPD ask participants to recall the previous 3 and 7 days, respectively. This weakness in the experimental design means it is difficult to identify whether the error is intrinsic to each prediction method or simply an artefact of the comparison between different days or time-periods.
In the absence of other suitable criterion free-living methods, researchers have encouraged the simple evaluation of the agreement and disagreement between measures . Previous studies have compared prediction methods to daily PA record scores over 7 days  and PAEE estimated from the adapted PA compendium over 24 h . Again the PASIPD was poorly correlated with the reference standard, whereas outputs from a tri-axial Actiwatch demonstrated a stronger correlation, r = 0.51 . Nightingale and colleagues  supported their earlier laboratory findings, demonstrating that the Actiheart™ with individual HR calibration explained more of the variance in free-living PAEE than using the Actiheart™ with proprietary algorithms. However, these analyses were only performed on a subsample of participants (n = 8) who had provided enough detailed information in PA logs to allow accurate estimation of PAEE using the adapted PA compendium for manual wheelchair users . This compendium only describes the energy cost of 63 wheelchair activities compared to the 821 specific activities included in the updated version of the compendium of physical activities for adults without disabilities . Consequently, coding of activities is less specific and accuracy of data is reliant on the quality of the self-report PA log. There are clear discrepancies between validating objective tools in a controlled-laboratory and free-living environments. Therefore, renewed efforts are required to validate measurement tools in both settings to determine convergent validity.
Statistical Approaches, Analytical Considerations and Future Directions
The majority of studies found strong associations between criterion measurements and outputs from wearable devices. However, in some instances where results from Bland Altman methods are also available, considerable random error has been reported [42, 66]. Where devices have been validated over a wide range of activities, of various intensities (in keeping with best practice guidelines ) a stronger correlation coefficient is likely. Consequently we encourage researchers to conduct multi-trait multi-method approaches , such as Bland Altman methods to assess agreement  or report measurement error . It is also important that authors are very clear about what error calculations have been performed and what error statistics are reported. Furthermore, it would be advisable for the wider academic community to produce a consensus statement addressing the clinical limits for PA/EE assessment error for devices used in this population.
It is possible that predicting EE/PAEE from linear regression equations may be too simple an approach to use in examining complex movements or behaviours . The activity protocols adopted by laboratory validation studies cited here mostly focus around wheelchair propulsion of various velocities. It is important to characterise this behaviour, as it will likely make a significant contribution to TDEE in free-living conditions (similar to ambulation in adults without disabilities). But as push frequency increases to match higher velocities, so too will accelerometer outputs. Therefore, whilst it might be appropriate to use linear regression methods to quantify PAEE associated with wheelchair propulsion, this approach might misclassify other types of physical activity. This is highlighted by considerable increases in measurement error for sedentary or atypical movements such as folding clothes [38, 43]. Greater error in more frequently performed low-intensity or sedentary behaviours has potentially considerable implications for the accurate determination of free-living EE in persons who use wheelchairs. A more ecologically valid approach would be to develop regression models based on a smorgasbord of activities common in the everyday lives of persons who use wheelchairs. It is possible that, by giving more weight to everyday activities (i.e. household chores or work-based tasks), such regression models may reduce estimation error.
An alternative solution to regression models would be to use new data analysis methodologies , including hidden Markov models , artificial neural networks [92, 93] and classification trees , which use the rich information to classify certain activities and derive a more accurate estimate of EE . To obtain such rich information, the shortest possible epoch (1 s) should be selected for activity monitor data collection , primarily to maximise the original PA related bio-signal being retained. Garcia-Masso et al.  recently developed and tested classification algorithms based on machine learning using accelerometers to identify specific activities performed by persons who use wheelchairs. This is encouraging since activity-specific EE algorithms developed for resting, wheelchair propulsion, arm-ergometry and deskwork can improve overall EE estimation [70, 97]. One important consideration that remains to be addressed is, how well objective measurement tools and associated algorithms capture elevated energy expenditure during recovery from MVPA (i.e. excess post-exercise oxygen consumption). It is conceivable that a physiological signal is required to accurately capture this information when acceleration signals post-exercise might be similar to resting values. Future research should consider; (i) applying and further developing new data analysis techniques, (ii) using more ecologically valid assessments that better resemble free-living conditions for persons that use a wheelchair and, (iii) evaluating the performance of EE prediction models during recovery after exercise (which contributes to TDEE).
Some of the principal limitations of previous validation studies are the relatively small sample sizes recruited, the mixed aetiologies for wheelchair use and, use of EE prediction algorithms without cross-validation. This is likely due to difficulties associated with recruiting from various disabled populations , and we encourage research groups to work collaboratively to recruit larger sample sizes. Using a diverse sample of participants and aetiologies for wheelchair use has been widely adopted [37, 39, 72, 83] and provides a robust model for the assessment of EE in the wider population of individuals who use wheelchairs, rather than a subgroup of that population. When the development of regression equations to predict EE and subsequent evaluation was conducted on the same sample of participants [37, 42], there is a tendency for the evaluation statistics to be biased and overly optimistic . Cross-validation is necessary, whereby the validity of developed algorithms are assessed using an independent sample of participants [54, 70]. We advocate employing a leave-one-out cross validation analysis  which has been employed previously [38, 43]. This permits an ‘independent’ assessment of EE prediction algorithms, and is an optimal approach when participant recruitment is particularly challenging.
Wearable Technology and Physical Activity Guidelines for Persons who use Wheelchairs
The American College of Sports Medicine (ACSM) have highlighted wearable technology as the top fitness trend for 2016 . Available consumer devices (Apple Watch, Microsoft Band, Fitbit Charge HR) are becoming increasingly sophisticated, incorporating multi-sensor technologies and are worn on the most appropriate anatomical location (wrist) to predict EE in persons who use wheelchairs. Apple recently announced at its Annual Worldwide Developer’s Conference that they have developed fitness tracking algorithms specifically for persons who use wheelchairs. Such wearable devices have the potential to provide wheelchair users with physical activity feedback which is informative and motivating . The feasibility of combining estimation methods should also be explored. Greater context regarding the location, type and purpose of physical activity behaviours are of huge importance in public health research. More detailed information may be achieved by combination of GPS and accelerometer outputs, especially when also incorporating self-report measures. This approach could help to understand specific personal and environmental barriers to exercise, which are numerous for persons who use wheelchairs .
It has been suggested that individuals with disabilities should strive to meet PA guidelines of 150 min of MVPA per week . These general population guidelines were informed by epidemiological evidence, using questionnaires, which capture the amount of activity required above normal lifestyle activities. While minutes per week represent an easy target for people to understand and attain, only the PARA-SCI and multi-sensor devices can currently be used in persons who use wheelchairs to generate total accumulated MVPA per day/week. Discrepancies have been shown between self-reported and objectively measured PA [104, 105]. Consequently, a recent review of data collected with accurate multi-sensor devices in adults without disabilities has suggested that ~1000 min per week of MVPA is a more appropriate target . To our knowledge, only one paper has attempted to establish MVPA cut-points for wrist worn accelerometer outputs in persons who use wheelchairs . However, accelerometer outputs alone (without complex data processing techniques) cannot easily detect the resistance of various movements that have similar acceleration profiles i.e. arm-crank exercise at 70 revolutions per minute; with no resistance (light-intensity activity) vs. 40 W (likely MVPA). Therefore, deriving MVPA cut-points for single unit wrist/arm accelerometers might have limited applicability, as direct outputs are unable to differentiate the resistance of certain arm movements (thus activity intensity) common in the everyday lives of persons who use wheelchairs. As such, measuring activity intensity is of utmost importance to accurately estimate MVPA, above and beyond daily PAEE/EE. Improvements in measurement techniques that capture this specific variable would significantly help to inform specific PA guidelines for persons with chronic disabilities who use wheelchairs.
There is now a renewed impetus to translate progress in measuring PA in adults without disabilities to persons who use wheelchairs, with the techniques reviewed here (i.e. self-report, physiological signals, accelerometry and multi-sensor devices), displaying varying degrees of success. Currently, selecting a PA assessment tool to use in this population presents a challenging proposition for clinicians and researchers alike due to differing outcome variables of interest, practicality/usability of the tool and population specific considerations. To help guide decision-making, Fig. 1 was developed to provide a systematic evaluation of the strengths and limitations of the different measurement tools reported herein. The PARA-SCI has been extensively developed and is the most suitable self-report measure to predict time spent performing various intensity activities. This methodology also captures the type of activities being performed, categorised as either LTPA or activities of daily living, which provide useful behavioural information. Tri-axial accelerometers worn on the wrist or arm are well tolerated and relatively unobtrusive . They offer a promising alternative to self-report methods for predicting PA/EE, particularly when combined with devices attached to the wheelchair or by incorporating complex data analysis methodologies. Multi-sensor devices, with algorithms developed specifically for the individual or generally for persons who use wheelchairs, demonstrate considerably improved error in the prediction of PA/EE during controlled laboratory protocols. It is possible that due to altered movement patterns and variations in metabolically active mass, predicting PA/EE in persons that use wheelchairs might be intrinsically more challenging. However, building on the current progress outlined in this review, we encourage the scientific community to rise to the challenge and provide innovative solutions to accurately predict free-living PA behaviours in this population. This is particularly important given the greater risk of non-communicable diseases, which are often associated with reduced activity, in persons with chronic physical disabilities who use wheelchairs.
Booth FW, Gordon SE, Carlson CJ, Hamilton MT. Waging war on modern chronic diseases: primary prevention through exercise biology. J Appl Physiol. 2000;88(2):774–87.
Haskell WL, Lee IM, Pate RR, Powell KE, Blair SN, Franklin BA, et al. Physical activity and public health: updated recommendation for adults from the American College of Sports Medicine and the American Heart Association. Med Sci Sports Exerc. 2007;39(8):1423–34.
Kesaniemi YA, Danforth E, Jensen MD, Kopelman PG, Lefebvre P, Reeder BA. Dose-response issues concerning physical activity and health: an evidence-based symposium. Med Sci Sports Exerc. 2001;33(6):S351–8.
Brage S, Brage N, Franks PW, Ekelund U, Wareham NJ. Reliability and validity of the combined heart rate and movement sensor Actiheart. Eur J Clin Nutr. 2005;59(4):561–70.
Collins EG, Gater D, Kiratli J, Butler J, Hanson K, Langbein WE. Energy cost of physical activities in persons with spinal cord injury. Med Sci Sports Exerc. 2010;42(4):691–700.
Conger SA, Bassett Jr DR. A compendium of energy costs of physical activities for individuals Who use manual wheelchairs. Adapt Phys Act Q. 2011;28(4):310–25.
Butte NF, Ekelund U, Westerterp KR. Assessing physical activity using wearable monitors: measures of physical activity. Med Sci Sports Exerc. 2012;44:S5–S12.
Dionyssiotis Y, Petropoulou K, Rapidi CA, Papagelopoulos P, Papaioannou N, Galanos A, et al. Body composition in paraplegic men. J Clin Densitom. 2008;11(3):437–43.
Biering-Sorensen B, Kristensen IB, Kjaer M, Biering-Sorensen F. Muscle after spinal cord injury. Muscle Nerve. 2009;40(4):499–519.
Schofield WN. Predicting basal metabolic rate, new standards and review of previous work. Hum Nutr Clin Nutr. 1985;39 Suppl 1:5–41.
Gomes AID, Vigario PD, Mainenti MRM, Ferreira MD, Ribeiro BG, Soares ED. Basal and resting metabolic rates of physically disabled adult subjects: a systematic review of controlled cross-sectional studies. Ann Nutr Metab. 2014;16(4):243–52.
Buchholz AC, Pencharz PB. Energy expenditure in chronic spinal cord injury. Curr Opin Clin Nutr Metab Care. 2004;7(6):635–9.
Landsberg L, Young JB, Leonard WR, Linsenmeier RA, Turek FW. Is obesity associated with lower body temperatures? Core temperature: a forgotten variable in energy balance. Metab-Clin Exp. 2009;58(6):871–6.
Martin Ginis KA, Latimer AE, Buchholz AC, Bray SR, Craven BC, Hayes KC, et al. Establishing evidence-based physical activity guidelines: methods for the Study of Health and Activity in People with Spinal Cord Injury (SHAPE SCI). Spinal Cord. 2007;46(3):216–21.
Buchholz AC, Martin Ginis KA, Bray SR, Craven BC, Hicks AL, Hayes KC, et al. Greater daily leisure time physical activity is associated with lower chronic disease risk in adults with spinal cord injury. Appl Physiol Nutr Metab. 2009;34(4):640–7.
Tudor-Locke CE, Myers AM. Challenges and opportunities for measuring physical activity in sedentary adults. Sports Med. 2001;31(2):91–100.
Shephard RJ. Limits to the measurement of habitual physical activity by questionnaires. Br J Sports Med. 2003;37(3):197–206.
Sallis JF, Saelens BE. Assessment of physical activity by self-report: status, limitations, and future directions. Res Q Exerc Sport. 2000;71(2):S1–S14.
Rimmer JH, Riley BB, Rubin SS. A new measure for assessing the physical activity behaviors of persons with disabilities and chronic health conditions: the Physical Activity and Disability Survey. Am J Health Promot. 2001;16(1):34–45.
Kayes NM, McPherson KM, Taylor D, Schluter PJ, Wilson BJK, Kolt GS. The Physical Activity and Disability Survey (PADS): reliability, validity and acceptability in people with multiple sclerosis. Clin Rehabil. 2007;21(7):628–39.
Kayes NM, Schluter PJ, McPherson KM, Taylor D, Kolt GS. The Physical Activity and Disability Survey—Revised (PADS-R): an evaluation of a measure of physical activity in people with chronic neurological conditions. Clin Rehabil. 2009;23(6):534–43.
Washburn RA, Zhu WM, McAuley E, Frogley M, Figoni SF. The physical activity scale for individuals with physical disabilities: development and evaluation. Arch Phys Med Rehabil. 2002;83(2):193–200.
Craig CL, Marshall AL, Sjostrom M, Bauman AE, Booth ML, Ainsworth BE, et al. International physical activity questionnaire: 12-country reliability and validity. Med Sci Sports Exerc. 2003;35(8):1381–95.
Ullrich PM, Spungen AM, Atkinson D, Bombardier CH, Chen Y, Erosa NA, et al. Activity and participation after spinal cord injury: state-of-the-art report. J Rehabil Res Dev. 2012;49(1):155–74.
Martin Ginis KA, Phang SH, Latimer AE, Arbour-Nicitopoulos KP. Reliability and Validity Tests of the Leisure Time Physical Activity Questionnaire for People With Spinal Cord Injury. Arch Phys Med Rehabil. 2012;93(4):677–82.
van der Ploeg HP, Streppel KRM, van der Beek AJ, van der Woude LHV, Vollenbroek-Hutten M, van Mechelen W. The physical activity scale for individuals with physical disabilities: test-retest reliability and comparison with an accelerometer. J Phys Act Health. 2007;4(1):96–100.
Martin Ginis KA, Latimer AE, Hicks AL, Craven BC. Development and evaluation of an activity measure for people with spinal cord injury. Med Sci Sports Exerc. 2005;37(7):1099–111.
Manns PJ, McCubbin JA, Williams DP. Fitness, inflammation, and the metabolic syndrome in men with paraplegia. Arch Phys Med Rehabil. 2005;86(6):1176–81.
de Groot S, van der Woude LHV, Niezen A, Smit CAJ, Post MWM. Evaluation of the physical activity scale for individuals with physical disabilities in people with spinal cord injury. Spinal Cord. 2010;48(7):542–7.
Wen CP, Wai JP, Tsai MK, Yang YC, Cheng TY, Lee MC, et al. Minimum amount of physical activity for reduced mortality and extended life expectancy: a prospective cohort study. Lancet. 2011;378(9798):1244–53.
Gebel K, Ding D, Chey T, Stamatakis E, Brown WJ, Bauman AE. Effect of moderate to vigorous physical activity on All-cause mortality in middle-aged and older Australians. JAMA Intern Med. 2015;175(6):970–7.
Samitz G, Egger M, Zwahlen M. Domains of physical activity and all-cause mortality: systematic review and dose-response meta-analysis of cohort studies. Int J Epidemiol. 2011;40(5):1382–400.
Lee M, Zhu WM, Hedrick B, Fernhall B. Determining metabolic equivalent values of physical activities for persons with paraplegia. Disabil Rehabil. 2010;32(4):336–43.
Lewis JE, Nash MS, Hamm LF, Martins SC, Groah SL. The relationship between perceived exertion and physiologic indicators of stress during graded arm exercise in persons with spinal cord injuries. Arch Phys Med Rehabil. 2007;88(9):1205–11.
Westerterp KR. Assessment of physical activity: a critical appraisal. Eur J Appl Physiol. 2009;105(6):823–8.
Chen KY, Janz KF, Zhu W, Brychta RJ. Redefining the roles of sensors in objective physical activity monitoring. Med Sci Sports Exerc. 2012;44:S13–23.
Washburn RA, Copay AG. Assessing physical activity during wheelchair pushing: validity of a portable accelerometer. Adapt Phys Act Q. 1999;16(3):290–9.
Garcia-Masso X, Serra-Ano P, Garcia-Raffi LM, Sanchez-Perez EA, Lopez-Pascual J, Gonzalez LM. Validation of the use of Actigraph GT3X accelerometers to estimate energy expenditure in full time manual wheelchair users with spinal cord injury. Spinal Cord. 2013;51(12):898–903.
Learmonth YC, Kinnett-Hopkins D, Rice IM, Dysterheft JL, Motl RW. Accelerometer output and its association with energy expenditure during manual wheelchair propulsion. Spinal Cord. 2016;54(2):110-114.
Hiremath SV, Ding D. Evaluation of activity monitors in manual wheelchair users with paraplegia. J Spinal Cord Med. 2011;34(1):110–7.
Kiuchi K, Inayama T, Muraoka Y, Ikemoto S, Uemura O, Mizuno K. Preliminary study for the assessment of physical activity using a triaxial accelerometer with a gyro sensor on the upper limbs of subjects with paraplegia driving a wheelchair on a treadmill. Spinal Cord. 2014;52(7):556–63.
Nightingale TE, Walhim JP, Thompson D, Bilzon JL. Predicting physical activity energy expenditure in manual wheelchair users. Med Sci Sports Exerc. 2014;46(9):1849–58.
Nightingale TE, Walhin JP, Thompson D, Bilzon JLJ. Influence of accelerometer type and placement on physical activity energy expenditure prediction in manual wheelchair users. PLoS One. 2015;10(5):e0126086.
Van Remoortel H, Giavedoni S, Raste Y, Burtin C, Louvaris Z, Gimeno-Santos E, et al. Validity of activity monitors in health and chronic disease: a systematic review. Int J Behav Nutr Phys Act. 2012;9:84.
Tolerico ML, Ding D, Cooper RA, Spaeth DM, Fitzgerald SG, Cooper R, et al. Assessing mobility characteristics and activity levels of manual wheelchair users. J Rehabil Res Dev. 2007;44(4):561.
Sonenblum SE, Sprigle S, Caspall J, Lopez R. Validation of an accelerometer-based method to measure the use of manual wheelchairs. Med Eng Phys. 2012;34(6):781–6.
Coulter EH, Dall PM, Rochester L, Hasler JP, Granat MH. Development and validation of a physical activity monitor for use on a wheelchair. Spinal Cord. 2011;49(3):445–50.
Fu JC, Liu T, Jones M, Qian G, Jan YK, Ieee. Characterization of wheelchair maneuvers based on noisy inertial sensor data: a preliminary study, 2014 36th Annual International Conference of the Ieee Engineering in Medicine and Biology Society (Embc). 2014. p. 1731–4.
Fu JC, Jones M, Liu T, Hao W, Yan YQ, Qian G, et al. A novel mobile-cloud system for capturing and analyzing wheelchair maneuvering data: a pilot study. Assist Technol. 2016;28(2):105–14.
Topol EJ, Steinhubl SR, Torkamani A. Digital medical tools and sensors. JAMA. 2015;313(4):353–4.
Conger SA, Scott SN, Bassett Jr DR. Predicting energy expenditure through hand rim propulsion power output in individuals who use wheelchairs. Br J Sports Med. 2014;48(13):1048–53.
Popp WL, Brogioli M, Leuenberger K, Albisser U, Frotzler A, Curt A, et al. A novel algorithm for detecting active propulsion in wheelchair users following spinal cord injury. Med Eng Phys. 2016;38(3):267–74.
Matthews CE, HagstrÖMer M, Pober DM, Bowles HR. Best practices for using physical activity monitors in population-based research. Med Sci Sports Exerc. 2012;44:S68–76.
Hiremath SV, Ding D. Regression equations for RT3 activity monitors to estimate energy expenditure in manual wheelchair users. 33rd Conf Proc IEEE Eng Med Biol Soc. 2011;7348-7351.
Bussmann HBJ, Reuvekamp PJ, Veltink PH, Martens WLJ, Stam HJ. Validity and reliability of measurements obtained with an “activity monitor” in people with and without a transtibial amputation. Phys Ther. 1998;78(9):989–98.
van den Berg-Emons RJ, Bussmann JB, Haisma JA, Sluis TA, van der Woude LH, Bergen MP, et al. A prospective study on physical activity levels after spinal cord injury during inpatient rehabilitation and the year after discharge. Arch Phys Med Rehabil. 2008;89(11):2094–101.
Postma K, van den Berg-Emons HJG, Bussmann JBJ, Sluis TAR, Bergen MP, Stam HJ. Validity of the detection of wheelchair propulsion as measured with an activity monitor in patients with spinal cord injury. Spinal Cord. 2005;43(9):550–7.
Bussmann JBJ, Kikkert MA, Sluis TAR, Bergen MP, Stam HJ, Vaccaro AR. Effect of wearing an activity monitor on the amount of daily manual wheelchair propulsion in persons with spinal cord injury. Spinal Cord. 2009;48(2):128–33.
Hiremath SV, Intille SS, Kelleher A, Cooper RA, Ding D. Detection of physical activities using a physical activity monitor system for wheelchair users. Med Eng Phys. 2015;37(1):68–76.
Hiremath SV, Intille SS, Kelleher A, Cooper RA, Ding D. Estimation of energy expenditure for wheelchair users using a physical activity monitoring system. Arch Phys Med Rehabil. 2016;97(7):1146–53. e1.
Kooijmans H, Horemans HLD, Stam HJ, Bussmann JBJ. Valid detection of self-propelled wheelchair driving with two accelerometers. Physiol Meas. 2014;35(11):2297–306.
Keytel LR, Goedecke JH, Noakes TD, Hiiloskorpi H, Laukkanen R, Van der Merwe L, et al. Prediction of energy expenditure from heart rate monitoring during submaximal exercise. J Sports Sci. 2005;23(3):289–97.
Luke A, Maki KC, Barkey N, Cooper R, McGee D. Simultaneous monitoring of heart rate and motion to assess energy expenditure. Med Sci Sports Exerc. 1997;29(1):144–8.
Achten J, Jeukendrup AE. Heart rate monitoring—applications and limitations. Sports Med. 2003;33(7):517–38.
Ainslie PN, Reilly T, Westerterp KR. Estimating human energy expenditure—a review of techniques with particular reference to doubly labelled water. Sports Med. 2003;33(9):683–98.
Tanhoffer RA, Tanhoffer AIP, Raymond J, Hills AP, Davis GM. Comparison of methods to assess energy expenditure and physical activity in people with spinal cord injury. J Spinal Cord Med. 2012;35(1):35–45.
Hayes AM, Myers JN, Ho M, Lee MY, Perkash I, Kiratli BJ. Heart rate as a predictor of energy expenditure in people with spinal cord injury. J Rehabil Res Dev. 2005;42(5):617–23.
Coutinho ACB, Neto FR, Beraldo PSS. Validity of heart rate indexes to assess wheeling efficiency in patients with spinal cord injuries. Spinal Cord. 2014;52(9):677–82.
Garcia-Masso X, Serra-Ano P, Garcia-Raffi L, Sanchez-Perez E, Giner-Pascual M, Gonzalez LM. Neural network for estimating energy expenditure in paraplegics from heart rate. Int J Sports Med. 2014;35(12):1037–43.
Hiremath SV, Ding D, Farringdon J, Cooper RA. Predicting energy expenditure of manual wheelchair users with spinal cord injury using a multisensor-based activity monitor. Arch Phys Med Rehabil. 2012;93(11):1937–43.
Tsang K, Hiremath SV, Cooper RA, Ding D. Evaluation of custom energy expenditure models for SenseWear armband in manual wheelchair users. J Rehabil Res Dev. 2015;52(7):793–804.
Conger SA, Scott SN, Flynn JI, Tyo BM, Bassett DR. Validity and accuracy of physical activity monitors for estimating energy expenditure during wheelchair locomotion. San Francisco: American College of Sports Medicine Annual Meeting; 2012.
Petrofsky JS. Thermoregulatory stress during rest and exercise in heat in patients with a spinal-cord injury. Eur J Appl Physiol Occup Physiol. 1992;64(6):503–7.
Brage S, Brage N, Franks PW, Ekelund U, Wong MY, Andersen LB, et al. Branched equation modeling of simultaneous accelerometry and heart rate monitoring improves estimate of directly measured physical activity energy expenditure. J Appl Physiol. 2004;96(1):343–51.
Nightingale TE, Walhin JP, Thompson D, Bilzon JLJ. Predicting physical activity energy expenditure in wheelchair users with a multisensor device. BMJ Open Sport ExercMed. 2015;1(1);0:e000008. doi:10.1136/bmjsem-2015-000008.
Villars C, Bergouignan A, Dugas J, Antoun E, Schoeller DA, Roth H, et al. Validity of combining heart rate and uniaxial acceleration to measure free-living physical activity energy expenditure in young men. J Appl Physiol (1985). 2012;113(11):1763–71.
Brage S, Westgate K, Franks PW, Stegle O, Wright A, Ekelund U, et al. Estimation of free-living energy expenditure by heart rate and movement sensing: a doubly-labelled water study. PLoS One. 2015;10(9):19.
Brage S, Ekelund U, Brage N, Hennings MA, Froberg K, Franks PW, et al. Hierarchy of individual calibration levels for heart rate and accelerometry to measure physical activity. J Appl Physiol. 2007;103(2):682–92.
Plasqui G, Westerterp KR. Physical activity assessment with accelerometers: an evaluation against doubly labeled water. Obesity. 2007;15(10):2371–9.
Bray GA. Energy expenditure using doubly labeled water: the unveiling of objective truth. Obes Res. 1997;5(1):71–7.
Tremblay JH, Peronnet F, Lavoie C, Massicotte D. Fuel selection during prolonged Arm and Leg exercise with C-13-glucose ingestion. Med Sci Sports Exerc. 2009;41(12):2151–7.
Jacobs KA, Burns P, Kressler J, Nash MS. Heavy reliance on carbohydrate across a wide range of exercise intensities during voluntary arm ergometry in persons with paraplegia. J Spinal Cord Med. 2013;36(5):427–35.
Warms CA, Whitney JD, Belza B. Measurement and description of physical activity in adult manual wheelchair users. Disabil Health J. 2008;1(4):236–44.
Ainsworth BE, Haskell WL, Herrmann SD, Meckes N, Bassett DR, Tudor-Locke C, et al. 2011 compendium of physical activities: a second update of codes and MET values. Med Sci Sports Exerc. 2011;43(8):1575–81.
Bassett Jr DR, Rowlands A, Trost SG. Calibration and validation of wearable monitors. Med Sci Sports Exerc. 2012;44(1 Suppl 1):S32–8.
Post MW. What to Do with “moderate” reliability and validity coefficients? Arch Phys Med Rehabil. 2016;97(7):1051–2.
Bland JM, Altman DG. Statistical methods for assessing agreement between two methods of clinical measurement. Int J Nurs Stud. 2010;47(8):931–6.
Staudenmayer J, Zhu W, Catellier DJ. Statistical considerations in the analysis of accelerometry-based activity monitor data. Med Sci Sports Exerc. 2012;44(1 Suppl 1):S61–7.
Strath SJ, Pfeiffer KA, Whitt-Glover MC. Accelerometer use with children, older adults, and adults with functional limitations. Med Sci Sports Exerc. 2012;44:S77–85.
Clark CCT, Barnes CM, Stratton G, McNarry MA, Mackintosh KA, Summers HD. A review of emerging analytical techniques for objective physical activity measurement in humans. Sports Med. 2017;47(3): 439-447.
Pober DM, Staudenmayer J, Raphael C, Freedson PS. Development of novel techniques to classify physical activity mode using accelerometers. Med Sci Sports Exerc. 2006;38(9):1626–34.
Staudenmayer J, Pober D, Crouter S, Bassett D, Freedson P. An artificial neural network to estimate physical activity energy expenditure and identify physical activity type from an accelerometer. J Appl Physiol. 2009;107(4):1300–7.
Trost SG, Wong WK, Pfeiffer KA, Zheng YL. Artificial neural networks to predict activity type and energy expenditure in youth. Med Sci Sports Exerc. 2012;44(9):1801–9.
Bonomi AG, Plasqui G, Goris AHC, Westerterp KR. Improving assessment of daily energy expenditure by identifying types of physical activity with a single accelerometer. J Appl Physiol. 2009;107(3):655–61.
Heil DP, Brage S, Rothney MP. Modeling physical activity outcomes from wearable monitors. Med Sci Sports Exerc. 2012;44:S50–60.
Garcia-Masso X, Serra-Ano P, Gonzalez LM, Ye-Lin Y, Prats-Boluda G, Garcia-Casado J. Identifying physical activity type in manual wheelchair users with spinal cord injury by means of accelerometers. Spinal Cord. 2015;53(10):772–7.
Hiremath SV, Ding D, Farringdon J, Vyas N, Cooper RA. Physical activity classification utilizing SenseWear activity monitor in manual wheelchair users with spinal cord injury. Spinal Cord. 2013;51(9):705–9.
Yilmaz DDCB. Recruitment of spinal cord injury patients to clinical trials: challenges and solutions. Top Spinal Cord Inj Rehabil. 2006;11(3):12–23.
Hastie Y, Tibshirani R, Friedman JH. The elements of statistical learning; data mining, inference, and prediction. 2nd ed. New York: Springer; 2009.
Thompson WR. Worldwide survey of fitness trends for 2016. 10th Anniversary Ed ACSM’s Health Fit J. 2015;19(6):9–18.
Western MJ, Peacock OJ, Stathi A, Thompson D. The understanding and interpretation of innovative technology-enabled multidimensional physical activity feedback in patients at risk of future chronic disease. PLoS One. 2015;10(5):13.
Rimmer JH, Schiller W, Chen MD. Effects of disability-associated Low energy expenditure deconditioning syndrome. Exerc Sport Sci Rev. 2012;40(1):22–9.
Nash MS, Cowan RE, Kressler J. Evidence-based and heuristic approaches for customization of care in cardiometabolic syndrome after spinal cord injury. J Spinal Cord Med. 2012;35(5):278–92.
Prince SA, Adamo KB, Hamel ME, Hardt J, Gorber SC, Tremblay M. A comparison of direct versus self-report measures for assessing physical activity in adults: a systematic review. Int J Behav Nutr Phys Act. 2008;5:56.
Steene-Johannessen J, Anderssen SA, Van der Ploeg HP, Hendriksen IJM, Donnelly AE, Brage S, et al. Are self-report measures able to define individuals as physically active or inactive? Med Sci Sports Exerc. 2016;48(2):235–44.
Thompson D, Batterham AM, Peacock OJ, Western MJ, Booso R. Feedback from physical activity monitors is not compatible with current recommendations: A recalibration study. Prev Med. 2016;91:389–94.
Learmonth YC, Kinnett-Hopkins D, Rice IM, Dysterheft JL, Motl RW. Accelerometer output and its association with energy expenditure during manual wheelchair propulsion. Spinal Cord. 2016;54(2):110–4.
Warms CA, Belza BL. Actigraphy as a measure of physical activity for wheelchair users with spinal cord injury. Nurs Res. 2004;53(2):136–43.
Conger SA, Scott SN, Fitzhugh EC, Thompson DL, Bassett DR. Validity of physical activity monitors for estimating energy expenditure during wheelchair propulsion. J Phys Act Health. 2015;12(11):1520–6.
The authors would like to thank Roger and Susan Whorrod, and the Medlock Charitable Trust for their kind donations to the DisAbility Sport and Health (DASH) research group at the University of Bath. The authors would like to acknowledge Dr Andrew Siddall for his review.
No financial support was received for the conduct of this study or preparation of this manuscript.
TEN conceived the review, which was revised with input from PCR, DT and JLJB. TEN conducted the literature review and drafted the manuscript with PCR also contributing to the manuscript preparation. The manuscript was critically revised for important intellectual content by DT and JLJB. All authors approved the final manuscript.
Tom Nightingale (TEN), Peter Rouse (PCR), Dylan Thompson (DT) and James Bilzon (JLJB) declare that they have no conflict of interest.
About this article
Cite this article
Nightingale, T.E., Rouse, P.C., Thompson, D. et al. Measurement of Physical Activity and Energy Expenditure in Wheelchair Users: Methods, Considerations and Future Directions. Sports Med - Open 3, 10 (2017). https://doi.org/10.1186/s40798-017-0077-0
- Physical Activity
- Spinal Cord Injury
- Rest Metabolic Rate
- Wheelchair User
- Physical Activity Energy Expenditure